|HONEY BEE SECTION: INNOVATIONS
|Year : 2018 | Volume
| Issue : 3 | Page : 155-158
Smart phone electrocardiogram – Bridging the gap
Bharath Raj Kidambi, Aayush Kumar Singal, Sai Krishna Reddy
Department of Cardiology, AIIMS, New Delhi, India
|Date of Web Publication||11-Jan-2019|
Dr. Bharath Raj Kidambi
Department of Cardiology, AIIMS, New Delhi
Source of Support: None, Conflict of Interest: None
The latest development in the area of medical technology and telemonitoring has been the smart-phone electrocardiogram. The wide availability of smart phones in India, makes it a potential handheld ECG machine. ECG is highly used investigation in cardiology, with its ability to pick up Acute coronary and arrhythmic events. In this article, we would discuss about two such devices SANKET life, which is made in India and the US FDA approved Kardia ALive Cor device.
Keywords: Coronary artery disease, diagnostic, widespread
|How to cite this article:|
Kidambi BR, Singal AK, Reddy SK. Smart phone electrocardiogram – Bridging the gap. J Pract Cardiovasc Sci 2018;4:155-8
| Introduction|| |
The 21st Century has witnessed a paradigm shift in technological advancement and a boom in mobile health technologies. Smartphone applications are becoming very common in the modern management of cardiovascular diseases (CVD). There is increasing emphasis on self-management and the use of mobile applications to improve the quality of health care in case of chronic diseases. The latest integration of technology and modern medicine is the smartphone electrocardiogram (ECG). Much of thefirst electrocardiograms since its inception by Willem Einthoven were bulky. Recent developments of smartphone devices have led to the miniaturization and integration of ECG into mobile phones. ECG has a wide variety of uses but mainly used for diagnosis of cardiac arrhythmia and myocardial infarction. In this review article, we will review the available smartphone ECG devices.
Cardiovascular disease in India
CVDs are the leading cause of death in India. Coronary artery disease (CAD) has rapidly increased in the younger and middle-aged population leading to a great loss in productivity. In Western populations, only 23% of CVD deaths occur before the age of 70 years; whereas in India it is around 52%. There has been a rapid epidemiological transition from infectious diseases to noncommunicable disease within a brief time. The overall burden of CVD in India is shown in [Table 1]. There have been poorer outcomes in rural areas due to lack of facilities to diagnose CVD in a timely manner.
The ECG is an essential noninvasive test, used for screening, diagnostic, and monitoring CVD. ECG machines have evolved from the bulky room-sized equipment to now wearable technology at present [Figure 1].
|Figure 1: Evolution of electrocardiogram machine – (a) the initial bulky galvanometer device discovered by Einthoven. (b) Conventional modern day portable electrocardiogram machine. (c) Handheld smartphone electrocardiogram device (SANKET LIFE).|
Click here to view
ECG is a highly used and reliable investigation in the field of cardiology, and there is a huge need of ECG in rural areas where even basic facilities may not be available. Hence, there is a drive to make ECG more compact and ubiquitous. The smartphone is an ideal technology that can be made to function like an ECG machine. It has an operating system, internet connection, easy touchscreen interface and has already widespread availability throughout India. In 2013, AliveCor Kardia was thefirst device in the USA to be FDA approved for recording ECG in medical applications in smartphones. Following that many companies have made their own versions.
Smartphone electrocardiogram devices
There are many devices available in the market. In India, Alive COR KARDIA, SANKET LIFE are two important medical device apps at present in the market. In the USA, there are other devices namely – ECG CHECK, D-Heart, Qardio Core, EPI Mini, and iHealth Rhythm. These devices operate on the same basic principle of a sensor lead which collects the signals and the transmission through Bluetooth or other methods to a smartphone for further data processing., The differences between the two devices available in India are shown in [Table 2]. We will be reviewing one of the products Sanket life in this article.
|Table 2: Specifications of the commonly available smartphone electrocardiogram devices|
Click here to view
It is approximately the size of a credit card. It can be easily fitted in the pocket.
There are three sensors. Two on the front panel of the device and one on the top to be used as a chest lead sensor [Figure 2].
The initial battery was chargeable with a USB cable provided, but subsequent designs have a replaceable watch battery inside which can be changed after its use.
12 leads electrocardiogram
Sanket life is able to record the electrical activity of all 12 leads but not synchronously. It has to be obtained serially. However, it does not use jelly or any actual electrodes. The actual procedure to obtain these 12 leads is shown [Figure 3].
|Figure 3: Chest sensor has to be placed in appropriate positions as the conventional precordial electrocardiogram leads.|
Click here to view
New algorithms are continuously being added, but the present device is capable of detecting atrial fibrillation and does the measurements of the intervals and segments. The accuracy of the device has to be validated in larger studies.
The PDF report generated is comparable to standard ECG paper, and it can be printed out or texted to the physicians immediately [Figure 4].
|Figure 4: Electrocardiogram obtained as PDF from Sanket life mobile app (a) and the corresponding electrocardiogram (b) from a conventional electrocardiogram machine in a patient with congenital heart disease.|
Click here to view
- Easy handheld device
- Reasonable price compared to ALIVE Cor which is valued at Rs. 14000 INR
- Can be used in the outpatient department, can be used by physician, technicians, and even by patients
- Works in a leadless wireless fashion
- Able to get all 12 leads for analysis
- Reasonable accuracy
- Stores images in Cloud Drive
- Able to provide a basic initial report
- They charge Rs. 175 rupees only for the expert cardiac opinion.
- The accuracy of the device has to be validated. Large studies need to be done with a comparison to a gold standard conventional ECG, before using the device in clinical practice. It has not been tried on pediatric patients, patients with pacemakers and congenital heart disease patients
- Although mentioned on their website as being able to take ECG through the shirt, it is not always possible to do so unless the shirt is a very loose piece of cloth. The data obtained from skin surface is better compared to one through the shirt
- It does not have an inbuilt data processor; you need a mobile phone with Bluetooth technology to use the device
- There are some variations in ECG obtained in winter and summer seasons, due to the fact of moisture of skin which acts as a conductor.
Studies using mobile applications
These devices used Kardia AliveCor for their studies.
- Assessment of Remote Heart Rhythm Sampling Using the AliveCor Heart Monitor to Screen for Atrial Fibrillation The REHEARSE-AF Study
- iTransmit study, a small (n = 60), single-center trial comparing conventional trans-telephonic monitoring to the Kardia Mobile device for monitoring patients for arrhythmia recurrence after the ablation procedure, the Kardia Mobile device had 100% sensitivity and 97% specificity in the detection of AF and atrial flutter
- iHEART study: A single-center randomized, controlled trial investigating the efficacy of a mHealth ECG technology intervention to improve the detection of atrial fibrillation
- ST LEUIS trial (ongoing) (Clinicaltrials.gov ID: NCT02498405), the authors showed that these reconstructed 12-lead ECGs from the Kardia Mobile device demonstrated excellent concordance with standard 12-lead ECGs for the detection of STEMI as well as non-STEMI.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Prabhakaran D, Jeemon P, Roy A. Cardiovascular diseases in India: Current epidemiology and future directions. Circulation 2016;133:1605-20.
Dimarco AD, Onwordi EN, Murphy CF, Walters EJ, Willis L, Mullan NJ, et al
. Diagnostic utility of real-time smartphone ECG in the initial investigation of palpitations. The British Journal of Cardiology 2018;25:20-30.
[Figure 1], [Figure 2], [Figure 3], [Figure 4]
[Table 1], [Table 2]